|Figure 1: Teledyne LeCroy's PP108,|
a representative passive probe
22 May 2014
15 May 2014
|Figure 1: S-matrices for one-, two-,|
and three-port RF networks
07 May 2014
In the first of two posts on how to control the Teledyne LeCroy HDO oscilloscopes, we cover a good deal of ground relative to the front-panel controls. One of the topics is the horizontal controls, which position and scale acquired waveforms horizontally on the instrument's display. In this short video, application engineer Jeff Krauss takes you through the application of the time/div and horizontal delay controls so as to best position your waveforms for optimal viewing.